SP801A Square Wave Pulser
The SP-801A produces negative high power square pulses with fast rise and fall times with extremely low output impedance (~4 Ω). The negative 400 V rectangular pulse is adjustable in 2 dB steps down to -50 V (-20 dB setting). This provides a simple, reproducible way of controlling the signal amplitude. Step-up transformers are optionally available for use with high impedance piezoelectric transducers or low-frequency EMATs to produce drive levels up to 1200 V. Additionally, specific matching networks for use with EMATs are also available.
Very high pulse power output
The low output impedance of approximately 4 Ω makes possible pulse powers of approximately 10 kW when working into a matched 4 Ω load. The average power output is limited to 4.5 W by circuitry which automatically shuts down the high voltage supply.
Pulse Width optimized for transducer frequency
The pulse width control for the SP-801A has eight positions, four of which are set for operating frequencies of 0.5, 1, 2.25 and 5 MHz. The remaining four positions allow continuous width control from less than 25 ns to greater than 14 µs. This makes it possible to adjust the width for half the period of any operating frequency from 50 kHz to greater than 20 MHz. Wider or narrower pulse widths (lower or higher frequency) available on special order; please contact us.
Internal or External Triggering
The internal repetition rate generator may be set from 0.17 Hz to 10 kHz in a 0.17, 0.25, 0.5, 1 ratio sequence through a 20 position rotary switch, or the unit may be triggered externally with either a positive or negative going TTL waveform. In some cases it may be convenient to trigger the unit from an existing spike pulser with spikes of up to 1 W of average power.
Square wave pulser and receiver system
The combination of Ritec’s SP-801A high power square wave pulser with the BR-640A broadband receiver constitutes the ideal instrumentation for driving EMAT measurements and other ultrasonic testing of high damping materials at low frequency.